<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 15:23:28
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item12.vue
 * @Description:
 *
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved.
-->
<template>
	<div>
		<table border="1" cellspacing="0" cellpadding="0" align="center" style="width: 100%;">
			<tr class="title">
        <td rowspan="3">序号</td>
        <td rowspan="3">检测项目</td>
        <td rowspan="3">检测方法与<br/>评价要求依<br/>据条款</td>
        <td rowspan="3">评价值</td>
        <td colspan="5">检测周期</td>
			</tr>
      <tr class="title">
        <td rowspan="2">验收检测</td>
        <td rowspan="2">状态检测</td>
        <td colspan="3">稳定性检测</td>
      </tr>
      <tr class="title">
        <td>日检</td>
        <td>月检</td>
        <td>年检</td>
      </tr>
			<tr>
				<td>1</td>
				<td>静态输出剂量</td>
				<td>6.1</td>
				<td>±2.0%内</td>
				<td>√</td>
				<td>√</td>
				<td>√</td>
				<td>√</td>
				<td>√</td>
			</tr>
      <tr>
      	<td>2</td>
      	<td>旋转输出剂量</td>
      	<td>6.2</td>
      	<td>±4.0%内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>√</td>
      	<td>√</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>3</td>
      	<td>射线质（百分深<br/>度剂量，PDD）</td>
      	<td>6.3</td>
      	<td>±3.0%内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>—</td>
      	<td>√</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>4</td>
      	<td>射野横向截面剂<br/>量分布</td>
      	<td>6.4</td>
      	<td>±3.0%内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>—</td>
      	<td>—</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>5</td>
      	<td>射野纵向截面剂<br/>量分布 </td>
      	<td>6.5</td>
      	<td>±1.0mm 内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>—</td>
      	<td>—</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>6</td>
      	<td>多叶准直器<br/>（MLC）横向偏移</td>
      	<td>6.6</td>
      	<td>±1.5mm 内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>—</td>
      	<td>—</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>7</td>
      	<td>绿激光灯指示虚<br/>拟等中心的准确<br/>性</td>
      	<td>6.7</td>
      	<td>±1.0mm 内</td>
      	<td>√</td>
      	<td>—</td>
      	<td>—</td>
      	<td>√</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>8</td>
      	<td>红激光灯指示准<br/>确性 </td>
      	<td>6.8</td>
      	<td>±1.0mm 内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>√</td>
      	<td>√</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>9</td>
      	<td>治疗床的移动准<br/>确性</td>
      	<td>6.9</td>
      	<td>±1.0mm 内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>—</td>
      	<td>√</td>
      	<td>√</td>
      </tr>
      <tr>
      	<td>10</td>
      	<td>床移动和机架旋<br/>转同步性</td>
      	<td>6.10</td>
      	<td>±1.0mm 内</td>
      	<td>√</td>
      	<td>√</td>
      	<td>—</td>
      	<td>—</td>
      	<td>√</td>
      </tr>
		</table>
	</div>
</template>

<script>
export default {
  name: "item43",
  data() {
    return {
    };
  },
  created() {
  },
  methods: {
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
</style>
